# Multimeter Working Principle | Type, Construction, Application

## What is Multimeter

A multimeter, also known as a volt-ohm meter or VOM, is a device used to measure various electrical parameters such as voltage, current, and resistance. It is a versatile instrument that can be used in electronics, automotive, and electrical industries, as well as in homes for troubleshooting electrical issues.

A typical multimeter consists of a display screen, a selection dial, and two probes (usually red and black) with which to make contact with the circuit being measured. Depending on the model, a multimeter can measure AC and DC voltage, AC and DC current, resistance, capacitance, frequency, and temperature. The selection dial is used to set the appropriate range and measurement mode for the specific parameter being measured.

Multimeters can be digital or analog, with digital ones being more common and providing higher accuracy and precision. They are an essential tool for anyone working with electrical circuits and systems, allowing them to diagnose and fix issues quickly and efficiently.

## Working Principle of Multimeter

The working principle of a multimeter is based on the fact that electrical parameters such as voltage, current, and resistance can be measured by measuring the flow of electrons through a circuit.

When measuring voltage, the multimeter is connected in parallel with the circuit, and it measures the potential difference between two points in the circuit. The voltage causes a current to flow through the multimeter, which is then converted into a voltage reading and displayed on the screen.

When measuring current, the multimeter is connected in series with the circuit, and it measures the flow of electrons through the circuit. The current causes a voltage drop across a resistor inside the multimeter, which is then converted into a current reading and displayed on the screen.

When measuring resistance, the multimeter applies a known voltage to the circuit and measures the resulting current flow. Using Ohm’s law (V=IR), the multimeter can calculate the resistance of the circuit and display it on the screen.

Modern multimeters are digital and use microprocessors to make measurements and display readings on the screen. They can also have additional features such as continuity testing, diode testing, and capacitance measurement.

## How multimeter work?

A multimeter works by using its probes to measure various electrical parameters in a circuit, such as voltage, current, and resistance. The probes are typically made of metal and have insulated handles to prevent the user from getting shocked.

To measure voltage, the multimeter is connected in parallel with the circuit being measured. The probes are touched to the two points in the circuit where the voltage is to be measured, and the voltage difference is displayed on the screen of the multimeter.

To measure current, the multimeter is connected in series with the circuit being measured. The probes are placed in series with the circuit, and the current flow is measured by the multimeter. This is done by measuring the voltage drop across a known resistance in the multimeter, which is then converted into a current reading and displayed on the screen.

To measure resistance, the multimeter applies a small known voltage to the circuit and measures the resulting current flow. The resistance can be calculated using Ohm’s law (R=V/I), where R is the resistance, V is the applied voltage, and I is the resulting current. The multimeter then displays the resistance on the screen.

Modern digital multimeters use microprocessors to make measurements and display the readings on the screen. They may also have additional features such as continuity testing, diode testing, and capacitance measurement.

## Components used in multimeter

The components used in a multimeter can vary depending on the type and model of the instrument, but some common components are:

Display: A multimeter typically has a digital or analog display that shows the measurement readings.

Probes: The probes are used to make contact with the circuit being measured. They typically have insulated handles and metal tips.

Selection dial: The selection dial is used to choose the measurement mode and range.

Voltage regulator: The voltage regulator ensures that the multimeter provides a constant voltage or current for measurement purposes.

Amplifier: The amplifier boosts the signal received from the probes to a level that can be measured and displayed.

Resistor: A resistor is used in the current measurement mode to create a known resistance in series with the circuit being measured.

Capacitor: A capacitor is used in the capacitance measurement mode to create a known capacitance and to filter out any AC noise from the measurement.

Microprocessor: Modern digital multimeters use microprocessors to make measurements, process data, and display readings on the screen.

Battery: A battery is used to power the multimeter.

These are some of the common components found in a multimeter, but the specific components and their configurations can vary depending on the model and features of the instrument.

## Type of multimeter

There are several types of multimeters available, including:

Analog multimeter: This is the traditional type of multimeter that uses a needle and a scale to display measurements. They are simple and inexpensive but are not as accurate as digital multimeters.

Digital multimeter (DMM): These multimeters use a digital display to show readings. They are more accurate and offer more features than analog multimeters, such as data logging and auto-ranging.

Clamp meter: This type of multimeter is used to measure current without breaking the circuit. It has a jaw-like clamp that can be opened and closed around a conductor to measure the current flowing through it.

Automotive multimeter: These are specialized multimeters designed for use in the automotive industry. They can measure parameters such as RPM, engine temperature, and battery voltage.

HVAC multimeter: These are multimeters designed for use in the heating, ventilation, and air conditioning (HVAC) industry. They can measure temperature, humidity, and other HVAC-specific parameters.

Benchtop multimeter: These are larger, more sophisticated multimeters that are designed for use in laboratories and other professional settings. They often have multiple inputs, high accuracy, and a wide range of features.

These are some of the most common types of multimeters available, but there are many other specialized types available as well. The choice of multimeter will depend on the specific needs of the user and the tasks they need to perform.

## Analog multimeter

An analog multimeter is a type of multimeter that uses a mechanical or analog display to show measurement readings. It works by using a moving coil meter that measures the current passing through it and moves a needle across a scale to display the reading.

Analog multimeters are simple and inexpensive, and they can measure various electrical parameters, including voltage, current, and resistance. They are also known for being more responsive to changes in the signal and can be easier to use for certain applications, such as checking for voltage fluctuations.

However, they are less accurate than digital multimeters and can be more difficult to read. They also require manual adjustment of the range and sensitivity, which can be time-consuming and error-prone.

Analog multimeters are still used in some applications, such as audio and radio-frequency (RF) testing, where their responsiveness to changes in the signal is beneficial. However, digital multimeters have largely replaced analog multimeters for most general-purpose measurements due to their greater accuracy, convenience, and ease of use.

## Working of Analog Multimeter

Analog multimeters work by using a moving coil meter to measure the electrical parameter being tested. The meter consists of a coil of wire that is suspended on a pivot and surrounded by a permanent magnet. When a current flows through the coil, it interacts with the magnetic field and causes the coil to move, which is then displayed as a reading on the meter scale.

To measure voltage, the analog multimeter is connected in parallel with the circuit being tested. The test leads are connected to the two points in the circuit where the voltage is to be measured. When the circuit is closed, a current flows through the meter’s internal resistance, causing the coil to move and the needle on the meter scale to deflect. The voltage reading can then be read from the scale.

To measure current, the analog multimeter is connected in series with the circuit being tested. The test leads are connected in series with the circuit so that the current flows through the meter. The amount of current flowing through the meter can be read from the scale.

To measure resistance, the analog multimeter uses an internal battery and a known resistance to create a voltage across the circuit being tested. The amount of current flowing through the circuit can then be measured, and the resistance can be calculated using Ohm’s law (R=V/I).

Analog multimeters are relatively simple and easy to use but are less accurate than digital multimeters. They can also be affected by external factors such as temperature and humidity, which can affect the accuracy of the reading.

## Analog Multimeter Measuring Quantities

Analog multimeters can measure several electrical quantities, including:

Voltage: Analog multimeters can measure AC and DC voltages. To measure voltage, the test leads are connected across the component or circuit being tested, and the voltage is displayed on the analog meter scale.

Current: Analog multimeters can measure current flowing through a circuit. To measure current, the multimeter is connected in series with the circuit, and the current is displayed on the meter scale.

Resistance: Analog multimeters can measure the resistance of a component or circuit. To measure resistance, the test leads are connected across the component or circuit, and the resistance is displayed on the meter scale.

Capacitance: Some analog multimeters can measure capacitance. To measure capacitance, the test leads are connected to the capacitor, and the capacitance is displayed on the meter scale.

Frequency: Some analog multimeters can measure the frequency of an AC signal. To measure frequency, the test leads are connected to the circuit, and the frequency is displayed on the meter scale.

Analog multimeters typically have several measurement ranges for each quantity, and the user must manually select the appropriate range before taking a measurement. The range selected should be high enough to prevent damage to the meter and low enough to provide an accurate measurement.

## Operation of Analog Multimeter

Analog multimeters operate by using a moving coil meter to indicate the value of the electrical quantity being measured. The meter consists of a coil of wire that is suspended on a pivot and surrounded by a permanent magnet. When a current flows through the coil, it interacts with the magnetic field and causes the coil to move, which is then displayed as a reading on the meter scale.

To measure a quantity, the test leads are connected to the multimeter and the circuit being tested. The multimeter can then be switched to the appropriate range for the quantity being measured. For example, to measure voltage, the multimeter is switched to the voltage range, and the test leads are connected across the component or circuit being tested.

When the circuit is closed, a current flows through the meter’s internal resistance, causing the coil to move and the needle on the meter scale to deflect. The voltage or current reading can then be read from the scale.

Analog multimeters typically have multiple measurement ranges for each quantity, which are selected manually by turning a range selector knob. The range selected should be high enough to prevent damage to the meter and low enough to provide an accurate measurement.

Analog multimeters are relatively simple to use but require the user to interpret the reading based on the position of the needle on the meter scale. They are less accurate than digital multimeters and can be affected by external factors such as temperature and humidity. However, they can be more responsive to changes in the signal and are still used in some applications, such as audio and radio-frequency (RF) testing.

Low cost: Analog multimeters are typically less expensive than digital multimeters, making them more accessible to hobbyists and beginners.
Responsiveness: Analog meters can respond more quickly to changes in the signal than digital meters. This makes them better suited for measuring rapidly changing signals such as audio and radio-frequency signals.
Simplicity: Analog multimeters are relatively simple to use, with only a few basic controls to adjust. They do not require batteries or external power sources to operate.
Durability: Analog multimeters are generally more durable and less likely to be damaged by overloading or mishandling.

Accuracy: Analog multimeters are less accurate than digital multimeters, particularly at low measurement values. They also have limited precision and resolution.
Interpretation: Reading the analog meter scale requires interpretation, which can be difficult for some users, particularly those with visual impairments or color blindness.
Environmental sensitivity: Analog multimeters are more sensitive to temperature, humidity, and other environmental factors, which can affect the accuracy of the readings.
Limited features: Analog multimeters typically have fewer features and measurement functions than digital multimeters, limiting their versatility for some applications.
In summary, analog multimeters are a simple and affordable tool for measuring electrical quantities, particularly for applications where speed and responsiveness are important. However, they are less accurate and have limited features compared to digital multimeters, making them less suitable for some applications.

## Digital multimete

A digital multimeter (DMM) is an electronic instrument used to measure several electrical quantities with high accuracy and resolution. Unlike an analog multimeter, a digital multimeter displays the measurement as a numerical value on a digital screen.

A digital multimeter typically has several measurement functions, including:

Voltage: Digital multimeters can measure AC and DC voltages with high accuracy and resolution.

Current: Digital multimeters can measure current flowing through a circuit. Some digital multimeters can measure both AC and DC currents.

Resistance: Digital multimeters can measure the resistance of a component or circuit with high accuracy and resolution.

Capacitance: Some digital multimeters can measure capacitance with high accuracy and resolution.

Frequency: Digital multimeters can measure the frequency of an AC signal with high accuracy and resolution.

Digital multimeters typically have several measurement ranges for each quantity, and the range is selected automatically or manually depending on the model. Digital multimeters may also include additional measurement functions such as temperature, diode testing, and continuity testing.

To measure a quantity, the test leads are connected to the digital multimeter and the circuit being tested. The multimeter is then switched to the appropriate range for the quantity being measured. The measurement value is displayed as a numerical value on the digital screen.

Digital multimeters are more accurate and precise than analog multimeters and are less affected by environmental factors such as temperature and humidity. They are also easier to read and interpret, particularly for users with visual impairments or color blindness. However, digital multimeters can be more expensive and may not respond as quickly to changes in the signal as analog multimeters.

## Working of Digital multimete

The working principle of a digital multimeter (DMM) is based on the use of an analog-to-digital converter (ADC). The ADC converts the analog voltage or current signal into a digital signal that can be processed by a microcontroller or digital signal processor (DSP).

The measurement process in a digital multimeter typically involves the following steps:

The test leads are connected to the multimeter and the circuit being tested.

The multimeter automatically or manually selects the appropriate measurement range for the quantity being measured.

The analog signal from the circuit is converted to a digital signal by the ADC.

The digital signal is processed by the microcontroller or DSP to calculate the measurement value.

The measurement value is displayed on the digital screen of the multimeter.

Digital multimeters typically have a high input impedance, which means that they draw very little current from the circuit being tested. This reduces the effect of the multimeter on the circuit and ensures accurate measurements.

Digital multimeters can also include additional features such as data logging, waveform capture, and Bluetooth connectivity for remote monitoring and control.

Overall, the use of digital technology in multimeters provides high accuracy, precision, and ease of use, making digital multimeters a popular choice for professionals and hobbyists alike.

## Operation of Digital multimete

The operation of a digital multimeter (DMM) typically involves the following steps:

Power on the DMM: Most digital multimeters have a power button or switch that needs to be turned on to start the device.

Select the measurement function: Depending on the quantity being measured, the user needs to select the appropriate measurement function using the rotary selector switch on the multimeter.

Select the measurement range: The user needs to select the appropriate measurement range for the quantity being measured. Most digital multimeters have an auto-ranging feature, which automatically selects the best range based on the input signal. Alternatively, some multimeters have manual ranging, which requires the user to select the range manually.

Connect the test leads: The test leads are connected to the multimeter and the circuit being tested. The black test lead is connected to the common (COM) terminal, and the red test lead is connected to the terminal for the quantity being measured, such as voltage (V) or current (A).

Take the measurement: Once the test leads are connected and the range and function are selected, the user can take the measurement by pressing the hold button or trigger.

Read the measurement value: The measurement value is displayed on the digital screen of the multimeter. Some multimeters may also display additional information, such as the measurement unit or decimal places.

Power off the DMM: Once the measurement is complete, the user can turn off the multimeter by pressing the power button or switch.

Digital multimeters may also include additional features, such as data logging, waveform capture, and Bluetooth connectivity, which require additional steps for operation.

High accuracy: Digital multimeters provide high accuracy and precision, with most models capable of measuring to within 0.1% of the reading. This makes them ideal for sensitive measurements where accuracy is critical.
Easy to read: The digital display of a DMM is easy to read, with most models featuring large, backlit displays that show the measurement value clearly and accurately.
Auto-ranging: Many digital multimeters have an auto-ranging feature that automatically selects the best range based on the input signal, making it easier to use for those who are less experienced with multimeters.
Multiple measurement functions: Digital multimeters can measure a wide range of electrical quantities, including voltage, current, resistance, capacitance, frequency, and temperature, making them versatile and suitable for a variety of applications.
Additional features: Digital multimeters often have additional features such as data logging, waveform capture, and Bluetooth connectivity, which can enhance their functionality and make them more useful for advanced users.

Cost: Digital multimeters are typically more expensive than analog multimeters, which can be a disadvantage for those on a tight budget.
Complexity: Digital multimeters are more complex than analog multimeters, with more features and functions that can be overwhelming for those who are less experienced with multimeters.
Battery life: Digital multimeters require a power source, typically a battery, which can run out of power and need to be replaced, potentially causing downtime during critical measurements.
Input impedance: While digital multimeters typically have a high input impedance, they can still draw some current from the circuit being measured, which can affect the accuracy of the measurement in some situations.
Overall, the advantages of digital multimeters, including their high accuracy, easy-to-read displays, and versatile measurement functions, make them a popular choice for professionals and hobbyists alike. However, the higher cost and complexity of digital multimeters may make them less suitable for casual users or those on a tight budget.

## Difference between Analog Multimeter and Digital Multimeter

Analog multimeters and digital multimeters have some significant differences, including:

Display: The most significant difference between analog and digital multimeters is the display. Analog multimeters use a needle or moving pointer to indicate the measured value, while digital multimeters use a digital screen to display the measurement value.

Accuracy: Digital multimeters offer higher accuracy and precision than analog multimeters. Analog multimeters typically have an accuracy of around 2-3%, while digital multimeters can measure to within 0.1% of the reading.

Range: Digital multimeters have a wider range of measurement capabilities than analog multimeters. Digital multimeters can measure a wider range of electrical quantities, including voltage, current, resistance, capacitance, frequency, and temperature, while analog multimeters are limited to measuring voltage, current, and resistance.

Ease of use: Digital multimeters are generally easier to use than analog multimeters, especially for less experienced users. Digital multimeters have auto-ranging capabilities that automatically select the best range for the input signal, while analog multimeters require the user to manually select the range.

Battery: Digital multimeters require a battery to power the digital display, while analog multimeters do not require any external power source.

Price: Analog multimeters are generally less expensive than digital multimeters.

In summary, digital multimeters offer greater accuracy, versatility, and ease of use, while analog multimeters are simpler, more affordable, and may be preferred by some users for certain applications.

## Application of Multimeter

Multimeters are versatile instruments that can be used in a variety of applications, including:

Electrical troubleshooting: Multimeters can be used to diagnose electrical problems in a wide range of devices and systems, including appliances, vehicles, and electrical installations.

Circuit testing: Multimeters can be used to test circuits and components, including resistors, capacitors, and diodes, to ensure that they are functioning correctly.

Power supply testing: Multimeters can be used to test the output of power supplies, including AC and DC voltage, current, and resistance.

Temperature measurement: Some multimeters have temperature measurement capabilities, making them useful for measuring the temperature of components or systems.

Automotive testing: Multimeters can be used in automotive applications to diagnose electrical problems, test batteries, and check the charging system.

Home electrical projects: Multimeters can be used for a variety of home electrical projects, such as checking electrical outlets, testing light switches, and diagnosing problems with appliances.

Educational purposes: Multimeters are often used in educational settings, such as in electronics courses, to teach students about basic electrical principles and to help them develop troubleshooting skills.

In summary, multimeters are versatile tools that can be used in a wide range of applications, from troubleshooting electrical problems to testing circuits and components, making them a valuable tool for electricians, technicians, hobbyists, and students alike.

## FAQ

What is the use of multimeter?
A digital multimeter is a test tool used to measure two or more electrical values—principally voltage (volts), current (amps) and resistance (ohms).

What are the two 2 types of multimeter?
Analog and Digital.

Is multimeter AC or DC?
Most multimeters can measure AC (Alternating current) as well as DC (Direct current) Current, but AC circuits can be dangerous.

You may also like